SIGVIEW includes a comprehensive set of triggering options available for data acquisition from
all supported devices. In this example, you will learn how to use these options to record multiple
repeating events and use these recordings for further analysis. The events you are trying to
record can be a hammer impact during vibration analysis, a sound from a musical instrument or
1. We will start by opening a Data acquisition dialog and performing some settings:
The device type, sampling rate, block length are not important - you can set those as needed for
"Start recording after trigger" options is turned on. This means that SIGVIEW will start recording
only after certain signal amplitude has been detected in the signal. The type of trigger is defined
by "Trigger on channel", "Threshold" and "Trigger type" options. In this example, SIGVIEW will
start recording if absolute signal level on Channel 1 increases over 50% of the maximal signal
amplitude (full scale).
You can also define offset for the recording start, relative to the trigger position. In this example,
we instruct SIGVIEW to start recording 1000 samples before the position where trigger was
detected. You can define positive or negative offsets, i.e. start recording before or after the
trigger. The maximal offset length is limited to the "Block length" number of samples.
After a trigger is detected in the incoming signal, SIGVIEW will start recording for "Recording
length" duration, in this example 2 seconds. You should set this parameter to be bigger then the
maximal duration of the event you would like to record.
"Re-Arm..." trigger option instructs SIGVIEW to start waiting for a new trigger as soon as one
recording has been made. After SIGVIEW detects a trigger for the first time and records 2
seconds of a signal it will automatically open new data acquisition window with the same settings
and will wait for another trigger in it. That way, each 2 seconds recording will remain in a separate
window. You can define maximal number of such windows with the "Max Re-Arm count".
2. Press OK in the data acquisition dialog and a signal window will open. Press "Rec" button in a
toolbar to start data acquisition, i.e. waiting for a trigger. Since there is still no signal, nothing will
happen immediately. In the background, SIGVIEW is acquiring your signal and testing if the
trigger has occurred.
3. Now, trigger the signal event by playing some sound, impacting with a hammer or anything
else causing the energy to be detected in the signal. SIGVIEW will start recording 1000 samples
before the event and the recording will last 2 seconds.
4. As soon as the recording has finished, a new window will open waiting for a next trigger.
Please repeat this 5 times to get 5 windows with recordings. Your SIGVIEW window should look
5. Now, use Window/Tile option to arrange all window. Select all signal windows in Control
Window (Context menu/Select all window) and choose Zoom-Out function in a toolbar to see the
complete content of all signal windows. The SIGVIEW window will look like this:
6. While still in Control Window, you can apply some analysis to all 5 recordings at the same time,
for example FFT. Also, you can show all recordings in an overlay window by using corresponding
options from a Control window's context menu.